EDT directION manufacture a range of conductivity calibration solutions to enable the successful and accurate calibration of our conductivity meters and electrodes.
The measurement of conductivity requires that the sample be aqueous and to be of adequate volume to cover the electrode surfaces. It is important that if the samples and standards are not of the same temperature then temperature compensation is required.
Conductivity increases approximately 2% per degree centigrade. Normally the conductivity calibration is carried out on one point. One useful check is that the conductivity in air should be zero as there is no conductive material between the electrodes.
EDT direction conductivity standards are Potassium Chloride solutions in either 100ml or 500ml sealed plastic bottles. For correct use the appropriate volume of standard should be poured into a clean beaker or container. Do not calibrate by placing the conductivity cell in the bottle as this will contaminate the solution. Following calibration remember to close the bottle to avoid evaporation or contamination and do not pour the used standard back into the bottle.
To view a simple calibration click here.
|Conductivity Calibration Solutions .pdf File Size 413.33KB Downloaded: 484|
|MSDS CONDUCTIV ITY STANDARD 12880 A3056.pdf File Size 1.37MB Downloaded: 783|
|Inner Seal||Yes. Transportable.|
Conductivity Calibration Solution 12.88mS/cm 500ml
- Brand: EDT directION Limited
- Product Code: A3056
- Availability: In Stock